Visit us on the web at for more information: • Price Quotations • Drivers· Technical Specifications. Manuals and Documentation . Model System DMM Instruction Manual Rev. D. Manual Type: Primary User; Part Number: D. Release Date: Sep 17 View and Download Keithley DMM instruction manual online. 5 function autoranging digital multimeter. DMM Measuring Instruments pdf manual.
|Published (Last):||5 January 2006|
|PDF File Size:||14.91 Mb|
|ePub File Size:||7.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the line frequency setting is changed but not saved: Connect the signal to be measured to the selected input terminals as shown in Figure Three time constants are used to allow the filters to settle. However, the window 19 of the internal filter are much smaller than the front panel filter window sizes.
The current reference level will be displayed. An error condition is also flagged in the serial poll byte, and the instrument can be programmed to generate an SRQ when an error condition occurs. To send the LLO command, the controller must perform the following steps: The only exception is the Write Enable WE pin, which are located at different positions on each chip.
The Model is then addressed to talk. Powering Results with Precision and Simplicity Guide: Storage will not occur. To minimize pick-up, keep the voltage source and the Model away from strong AC magnetic sources.
For 2-terminal measurements connect the resistance to the instrument as shown in Figure Translator error messages are listed and described in Table Press PRGM, 3, 1. Since an internal pull-up resistor is used, a mechanical switch may be used. If both the internal filter and the front panel filter are in use, the time constant is the sum of both filter values.
Digital filtering must be used to obtain a stable reading. Model repair forum thread EEVBlog: Of course, you can use pretty much any micro-controllers that you are familiar with as long as TTL logic levels are supported.
KEITHLEY DMM INSTRUCTION MANUAL Pdf Download.
IEEE interface functions are marked below the connector. The Model filters the signd by taking the weighted average of a number of successive reading samples.
Digital Calibration— When performing digital calibration, two points must be calibrated on each range. Table liste the contact assignments for the bus, and Figure shows the contact configuration.
The instrument will return to the operating states set up in step 1. There are certain words keitnley characters that cannot be used as defined Translator words. Has an internal cal enable jumper.
Keithley 196 DMM Instruction Manual
To exit from manuall menu, press the PRGM button. Once triggered, the instru- ment will perform the measurement sequence in accord- ance with previously selected rate and sample parameters.
This feature allows for zero correction or keithlsy of baseline values. Verification may also be performed whenever there is a question of instru- ment accuracy, or following calibration, if desired.
The current zero value will then be displayed. To use the Manuao with the Modelproceed as follows: Tne other secondary function is to enter the number 8. For exam- ple, if the multiplexer is on, the following message will be displayed: Voltmeter Complete Pulse Specifications Figure These commands are covered in detail in the following paragraphs.
Set trigger mode, and storage parameters.
HP Programming Example — Place the instrument in the remote mode with the following statement: Leithley maximum cable length is 20 meters, or 2 meters times the number of devices, which ever is less. SDC will return the instrument to the default line fre- quency setting.