Find great deals for Endress Hauser Prosonic FMU Ultrasonic Transmitter Measuring System. Shop with confidence on eBay!. matrix fmu – Download as PDF File .pdf), Text File .txt) or read online. FMU Ultrasonic Measurement Operating Instructions. FMU FMU FMU Endress + Hauser The Power of Know How.
|Published (Last):||28 March 2010|
|PDF File Size:||20.72 Mb|
|ePub File Size:||12.19 Mb|
|Price:||Free* [*Free Regsitration Required]|
The length unit is the same as that before the reset Any curve which has been previously entered by the user remains stored;the transmitter selects the linear mode.
Endress Hauser Prosonic FMU Ultrasonic Transmitter Measuring System | eBay
The Prosonic FMU transmitter 680 be installed outside the explosion hazardousarea. Post on Dec views. In V0H9 the level can be read V4H9 for channel 2. The board with the transmitterelectronics is mounted separatelyin the control cabinet. Ensure that the power is switched off when the unit is connected up. Only the software totalizer can be reset. The Prosonic FMU or has a current output for Configuration of thecyclic data telegramUse the configuration software of your PLC in order to compose the data telegram from these modules in one of the following ways: Theselevel applications are given in Appendix B.
A green LED also indicates that operation is free from faultsand flashes on warning. In a Venturi channel, the flowrate is measured at its optimum, if the ratio h2 downstream water to h1 head water does not exceed the factor 0.
Manual Prosonic Fmu 860
After entering the point number the asigned point of values, level and flow, can bedisplayed The more pairs of values you enter, the more exact the linearization will be. Prosonic FMU for flow measurement in channels and weirs, Prosonic FMU the two-channel version for the measurement of flow rate or level with the first channel and level measurementonly with the second channel or differential or average measurement.
Valid device addresses are in the range from 1 to EmptyThis module must be applied during configuration see belowif main value 2 is not to appear in the data telegram. When measuring the difference or average, channel 1 shows the unchanged measurandof the sensor.
QV 40,09 1. Note the settings When entering the parameters the values entered can be noted in the table on page The certified sensor ensures that the measuring system can be usedin explosion hazardous areas. Levels coming within the blocking distance can cause theinstrument to malfunction.
V9H3 Unit code with software version The first two figures are the unit code, the lasttwo are the software numberTab. Press H to select the horizontal position Press simultaneously to select the measured value field, V0H0Parameter entry Select the digit to be changed. The parameters are set usingsubmenus.
Loosen the metallic braiding, twist it together securely and connect it to terminal 80 Channel 1 or terminal 90 Channel 2 Fig. To shorten the time taken for commissioning, for levelmeasurement, an application parameter can be selected which automaticallysets the ultrasonic measuring line to one of five different applications.
With the two-channel unit FMU the changeover from Units of length are the same after resetting the Prosonic as before the resetSetting the operatingmode Now enter a number in V8H0 for the operating mode: The modes 7 and 8, Simulation channel 1 and channel 2, are described in chapter 9.
If linearization is subsequently performed in a different technical unit, all settingshave to be altered for all relays acting as limit switches. Exceeding the level tocome within the blocking distance of the sensor is also indicated promptly by the display,the signal output and the relays.
Manual Prosonic Fmu – [PDF Document]
Each relay is assigned a number which has to be selected before entering the relayfunction. The sensors are then scanned in sequence.
IP 66 protective housing. FDU 83, 84, 85, It is divided into the followingsections: