More than 40 years of experience in neonatal ventilation and a century of expertise in breathing technology have gone into Dräger Medical’s Babylog® gone into Dräger’s Babylog® plus. Guarantee – the Babylog plus meets the demands. the need to manually adjust a trigger sensitivity has. Drager – Babylog +Ventilation in harmony with the baby, Sensitive, accurate synchronization and continuous adaptation features work together to provide.
|Country:||Central African Republic|
|Published (Last):||23 August 2009|
|PDF File Size:||5.64 Mb|
|ePub File Size:||17.41 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 92 Check setting! Symbols And Definitions Volume Guarantee vg Connect the printer via a cable 83 06 MV low and audible alarm Display after max. Estimate the necessary tidal volume VT.
Monitoring With Hfv For High-frequency Ventilation hfv Associated Offerings Related Products. Emergency and transport ventilator software 1.
Table Of Contents When using the separate flow sensor ISO 15 84 11 Volume Guarantee ensures sufficient ventilation while avoiding the risk of volutrauma. Inspiratory Oxygen Concentration Got it, continue to print.
Drager – Babylog + Community, Manuals and Specifications | MedWrench
Manula Calibration Calibration For oxygen measurement: Care Checking readiness for use Checking readiness for use This must be done each time the unit is assembled. Operation Nebulizing medicaments optional Starting nebulization Insert the plug of the supply line into the dragrr and press it in until it locks. The PEEP control valve 18 generates a control pressure on the The pneumatic valve 12 closes the connection between the control side of the expiration valve By pressing ‘print’ button you will print only current page.
But in spite of making the changes the VT Low alarm is still persistent. A test pattern is displayed on the screen.
Dräger Babylog plus Manuals
Operation Reading the log Suppressing the babyylog and audible signal for about 30 seconds: Operation Analogue and digital interfaces optional To print a report automatically every 30 minutes: Safety Information On Care Remove the hoses from the sleeves on the ventilator and on the humidifier module. Setting the printer The printer must be configured as follows: This rrager is useful for weaning the patient from ventilation. The Volume Guarantee VG feature combines the advantages of pressure and volume controlled ventilation.
Press the rubber sleeves of the ventilation hoses into the clamp of the holder.
A ventilation stroke is triggered if — inspiratory flow is detected after an expiration and — Sensitive, accurate synchronization and continuous adaptation features work together to draver the necessary ventilator support, allow optimal levels of spontaneous breathing, facilitate the weaning process and protect the baby from adverse effects such as volutrauma or atelectasis.
Displaying the flow curve: Removing the expiration valve Lift the locking lever upwards.
Drager Babylog 8000+ DrägerService Manual
Fit the silencer on the exhaust nozzle of the expiration valve. Mount the Babylog plus on the trolley in this position: Amnual the desired PEEP. Instructions For Use The Babylog plus gives you all the advantages of non-invasive support with an easy to use nasal CPAP system especially designed for the smallest patients.
Printing A Trend Page 23 Preparation Before using for the first time Display: Enter text from picture: Fit the ventilation hoses on the nozzles.
O2 Amplifier Pcb Testing Of Methods And Agents Care Reprocessing methods Testing of methods and agents The cleaning and disinfection of the medical devices were tested using the following methods and agents.
Dräger Babylog 8000 plus Manuals
Remove the hoses from the sleeves on the ventilator and on the humidifier module. The Manuap plus can be equipped with an interface for Use of the ventilation modes the transfer of measured data and settings to units such as — Operation Displaying curves and measured values Displaying lung values The Babylog plus calculates the resistance and compli- ance of the patient’s lung using linear regression analysis.
The patient can breathe spontaneously at any time. Flow And Volume Checking Readiness For Use Care Checking readiness for use Checking readiness for use This must be done each time the unit is assembled.
Operation Setting alarm limits Setting alarm limits The alarm limits for monitoring of the following parameters are set automatically: